DATEL, Inc., Mansfield, MA 02048 (USA) • T el: (508)339-3000, (800)233-2765 Fax: (508)339-6356 • Email: sales@http://m.wendangku.net/doc/b0387cde195f312b3169a528.html • Internet: http://m.wendangku.net/doc/b0387cde195f312b3169a528.html For DM-3000 and DM-4000Series Panel Meters
Part No. RN-3100/4100PURPOSE
This kit contains all the components necessary to modify DATEL panel meters such that they accept signal inputs other than the standard ±2.0V range. This kit will modify the following panel meter models:
DM-3100X -1DM-4105-1DM-31003-1DM-3104 -1DM-4200-1
Using the components in this kit, these panel meter models can be modified to accept any one of the following signal ranges:
There is one exception: 4½ DIGIT PANEL METERS (ANY OF THE MODELS WHOSE SUFFIX BEGINS WITH A “4”) CANNOT BE MODIFIED FOR THE ±200mV RANGE.
MAKING THE MODIFICATION
Making a range change modification is a relatively simply procedure, however, familiarity with proper soldering techniques is required. When handling the panel meter printed circuit boards, care should be taken to avoid a dis-charge of static electricity which could damage the meter’s sensitive CMOS circuits. A grounding wrist strap should be worn when handling the PC boards.
To make a range change modification, follow these steps:
1.Locate your panel meter model and the desired range in the tables below. Note that the 200mV range is the only range which requires different components depending on the meter model.
INNOVATION and C E E N
O b s o l e t e P r o d u c t N o t a v a i l a b l e f o r s a l e